Time: 2021 Jul 25 1:04:25
Found 2637 logbook entries, displaying most recent 100
Logbook Contents (Most Recent Entries)
21 Jul 2021 Wednesday 12:06 [Leblanc] Shutdown and inspected alphatross source. Found Rb contamination throughout the plasma bottle. Cleaned and reassembled source. Reduced LOBS loop temperature from 40 to 35 centigrade.
21 Jul 2021 Wednesday 09:05 [Leblanc, Carter] Alphatross probe short Monday afternoon. Source turned off on Monday. Tuesday, source backfilled and bottle removed. Found dark grey deposits on the inside of the plasma bottle, which disappeared on exposure to air. Bottle cleaned, canal and BN insulator replaced, vacuum restored. Current unstable at first, but eventually nearly 1uA in the LE cup, fairly stable. Left source running overnight. Wednesday morning, no beam current. Found probe had developed a short starting around 4 AM. Short was intermittent until around 7, at which point it became steady and pulled the probe voltage down to around 1kV.
08 Jul 2021 Thursday 10:28 [Tom Massey] 7/8/2021 Operator DRD reset 0.42 mr reading.
21 Jun 2021 Monday 23:47 [Yenuel Jones-Alberty] Target current sharply increased at around 10:30 pm tonight, and then plummeted to zero. Control milliampere light on Target Glasmann was then observed to be on. We swapped the deuterium pellet for a fresh pellet, and are currently turning the source back on. The old pellet appears to still have some titanium deuteride.
17 Jun 2021 Thursday 10:12 [Tom Massey] Survey for 9Be(d,n) reaction on swinger 30 nA beam. No detectable neutrons or gamma at Small Target Room Door. Larget targe Room 25 cts neutrons at 15 R beam leg opening no gamma detected. Garage area 25 cts thermal and 50 cts fast neutrons at door wheel wells, no gamma.
14 Jun 2021 Monday 09:56 [Leblanc] Duterium pellet removed from source. Pellet appeared depleted. New pellet installed. Turning on HV supplies causes target PSU to go into current limiting mode. Need to service the source.
11 Jun 2021 Friday 18:20 [Tom Massey] The target high voltage shorted out and is now in current control mode. I turned the target voltage down and this did not change the current reading. I shut the source down and the target current dropped when the ionizer was turned down. The short came back when the target current was turned on and stayed on. The source has been shut down.
10 Jun 2021 Thursday 16:06 [Carter/LeBlanc] The new AMI Oxygen Deficiency monitors are installed and calibrated, one on the dock and one in the pit. The the display value for normal oxygen level will read 20.9% locally on these monitors. A green LED on the front of the unit indicates normal safe condition. If the level drops to 20.0% the status LED will change from green to yellow. Then, if the level drops to below 19.5%, the status LED will turn red and a 95dB local alarm will sound. Pressing "Silence" will silence the alarm for one minute. Leave the area immediately and call for help. If the status LED is flashing red, AC power has been lost and the unit will continue to operate for about one hour. The oxygen sensor data is logged and a plot available on the real-time data plots under the "Miscellaneous" section labeled "Pit_O2" and "Dock_O2".
04 Jun 2021 Friday 16:34 [Leblanc] Alphatross source cleaned and rubidium charge replaced. Pumped to high vacuum until about 2PM today, then started source. Obtained approximately 5uA of current at the source cup, but only 9nA at the LE cup. Turned Rb oven off, letting plasma run over the weekend.
04 Jun 2021 Friday 16:21 [Carter/LeBlanc] Remote logging of the vault pit O2 monitor connected and functioning. O2 levels and alarm status can be viewed in the control room for the vault pit O2 monitor.
04 Jun 2021 Friday 11:59 [Carter/LeBlanc] New AMI Model 221R O2 oxygen deficiency monitors installed and functional in local mode. Remote mode communications and software has been tested but wires will need to be run for remote monitoring connections to the control room computer. One unit is in the accelerator pit and one on the accelerator dock area and have local readouts and alarms for the respective areas.
01 Jun 2021 Tuesday 17:56 [Greg] Trouble with the alphatross source today. Plasma color looks purple/pink on the monitor, and beam current is very low. Probe current appears within normal range. Current PV 6kV, PC 3mA, EV 5.5kV, Oven at 80V, Bias at 49kV. Chamber temp is 49 Celsius, oven temp 172 Celsius, 0.2 SCCM He4. Resulting negative beam current of 6.5nA. Disconnected probe leads and used megger to test for current through the plasma bottle, no current, no plasma induced. Back filled source with argon and checked for and found Ar flow through the He inlet. Roughed and pumped source down, plasma color still off, current still low. Turning off source, leaving "water" on overnight to try to keep the chamber warm for the morning. Need to determine next steps.
17 May 2021 Monday 17:25 [Greg, Tom, Nisha] Spent time today checking telescope alignment at swinger. Carefully set height using movable device, and checked alignment on fixed telescope. Made SMALL adjustment to level and yaw on fixed telescope. Alignment is complete the best of our ability given current tooling. Inscribed marks on the back of the tunnel appear to be correct.
15 May 2021 Saturday 18:21 [Tom Massey] I spilled some water near the oxygen sensor in the Garage area. The system gave an alert several minutes afterwards. It shows as trouble on the pannel.
13 May 2021 Thursday 11:09 [Carter/LeBlanc] Problem with analyzing magnet polarity reversal was diagnosed and repaired. Mechanical obstruction from drooping high current magnet wire was interfering with the contact mechanism. Obstructing wire moved and secured away from the contact.
06 May 2021 Thursday 11:09 [Greg, Yenuel] Came in this morning and inspected the pellet. Looked "icky", see https://photos.app.goo.gl/tWauxzzaVLHh5csv5 Made new pellet, loaded into sputter source. Short seems to have cleared, new pellet is warming up now.
05 May 2021 Wednesday 20:55 [Justin Warren] Update Dropping the cesium resivoir current did not reduce the target current after ~30 minutes and beam was not regained on the low energy cup. Proceeded to shut down the sputter source and accelerator. During the shutdown of the sputter source, while dropping the ionizer current, the target current dropped to 0 mA.
05 May 2021 Wednesday 20:06 [Justin Warren] At approximately 7:30 PM on 05/5/21, the target current on the sputter source reached ~21 mA, and the target voltage went to current control mode. The target voltage and resivoir current were reduced to 0 to attempt to fix the issue. Further update in 30 minutes.
05 May 2021 Wednesday 09:31 [csrter] Just documenting that the sputter source oven heater is 115 VAC 200W with a cold resistance of 75 ohms.
05 May 2021 Wednesday 08:36 [Tom Massey] Debugging the sputter source. The cesium resivoir heater was checked. It checked good but was replaced. The resistances on the sputter source acceleration tube were checked. The proton pellet was replaced with an old deuterium pellet. This pellet gave 6 microamps in two hours. A new proton pellet was made. This pellet was placed in the bell jar and the pressure reduced to roughing vacuum. The bell jar was backfilled with dry nitrogen. This pellet gave 2 microampers of beam in under a half hour. The temperature on the cesium oven was raised from 117 to 120 degrees. This gave a factor of ~4 increase in beam current on target.
04 May 2021 Tuesday 09:29 [Greg] Low current from sputter source yesterday. After some testing, we decided to check cesium level, found reservoir empty. Refilled with 1g of Cesium, restarted source to run overnight. Beam current still low this morning, still trying to troubleshoot.
29 Apr 2021 Thursday 11:02 [carter] I have replaced the suspect component causing instabilities in the voltage display of the sputter source extractor supply. Testing did not reveal the previous symptoms of instability. As of this time you can believe the voltage reading unless symptoms re-appear. If you see instabilities we previously saw, let me know and go back to the voltage set dial for setting the supply. As a rough approximation, 16 KV per dial revolution of the voltage set.
28 Apr 2021 Wednesday 11:45 [carter] When moving the swinger, watch the cables and hoses near the rotating seal end (beam injection side) and make sure they don't catch on anything as the swinger moves. A temporary repair has been made on the damaged cable and will be made more permanent when time permits.
28 Apr 2021 Wednesday 11:20 [carter] As mentioned by Tom in the Monday lab meeting, the sputter source extraction power supply voltage meter seems to be unstable. The supply however seems to be working OK and stable, so until further notice, use the calibrated voltage set dial to set the voltage. The supply output voltage = (control dial setting / 10) * 160KV. Dial setting = (desired voltage / 160) * 10. The dial is a 10 turn dial so the integer is the displayed number and the fraction is read from the dial. Example for 58 KV, (58 / 160) * 10 = 3.625 on the voltage set dial.
28 Apr 2021 Wednesday 10:48 [carter/massey/subedi] Roughing pump control cable had snagged when swinger was rotated and mechanically damaged. Repaired and pump now runs. The ion gauge was leaking between the 2 swinger magnets and was replaced. Swinger now pumping down.
27 Apr 2021 Tuesday 23:23 [Tom Massey] The turbo pump controller shows 0 RPM and an erorr code of E006. The turbo pump's roughing pump was not running and was cold when the turbo pump was on.
27 Apr 2021 Tuesday 23:10 [Tom Massey] The swinger was moved to zero degrees. The vacuum interlock between the target and turbo section of the swinger shut the valve. The beamline gate valve also closed. I was called ot check the status. The turbo roughing vacuum read atmosphere when I arrived. There was good vacuum yet in the turbo section of the swinger. There was good roughing vacuum in the target section. We were unable to open the valve between the target and turbo section. The turbo pump was turned off for safety as the roughing vacuum pumb was not running. The ion guage was bumped in the debugging of the problem and it shorted out.
26 Apr 2021 Monday 12:13 [Tom Massey] Conditioned the tandem to 4.6 MeV on GVM. This went smootly with no large outgasing. This was done at 50 keV per each 5 minutes from 4.0 MeV GVM. The Dell high voltage supply had been showing swings of voltage from 37-42 keV on the display. The current is very stable. This may indicate a problem with the gauge rather than with the stablity of the system. However, the actual voltage is not clear. Manual programing shows 3.2 on the dial.
15 Apr 2021 Thursday 15:30 [Tom Massey] We began running 8 MeV alphas. This gave a analyzer magnet current of 93.9 Amps. During tuning and at the begining of the run the current increased several times up to 94.1 Amps. This caused the loss of the beam. It could be recovered by adjustment of the switcher current. After 3 hours the magnet current settled down, the swings of current are now absent.
05 Apr 2021 Monday 11:13 [Greg] Tom, Yenuel, and Doug found bad vacuum on the analyzer leg this morning. I investigated a little and found rough vacuum was OK, but gate valve on the cryo pump didn't want to open. Bypassed interlock and let it pump down for a while. Vacuum all seems OK at this point, though not as low as normal operating (probably just needs time). Opened the nearby gate valves to maximize pumping speed. Vacuum is low enough for Doug to tune alphas for UGL.
23 Mar 2021 Tuesday 14:26 [Greg] 4 bottles of SF6 (total weight, approximately 460 lbs, 210kg) transferred into storage tank. Current storage tank pressure is approximately 25 PSI. This should allow us to increase the pressure in the tank by as much as 16PSI. Current pressure is running about 79 PSI. I think the new SF6 should be dried before being added to the accelerator, which will require about 3-4 days without running the accelerator. We should attempt to do this before running at the full 4.5MV, though the runs at 3.5MV have shown no signs of insufficient insulation.
22 Mar 2021 Monday 17:36 [Don, Greg] Buncher 4-65A output tube drawing grid current, shutting down at 1.2 KV. Replaced with new tube and now operating normally again. Note: the first replacement tube (NOS) was defective producing no grid drive so another NOS tube was installed.
22 Mar 2021 Monday 09:40 [Greg] Alphatross interlock controller had a blown fuse 1A/250V fuse. Replaced the fuse and was able to start normally. Will write up troubleshooting instructions.
22 Mar 2021 Monday 09:05 [carter] BCI counter 10's digit repaired.
21 Mar 2021 Sunday 16:28 [Ingram] LE cyropump working normally. 10L scattering chamber vacuum looks okay now the ion gauge has been outgassed. Alphatross vacuum still shutdown.
21 Mar 2021 Sunday 12:54 [Ingram] LE cryopumps did not start properly, iced up and not at temperature. Have warmed them for three hours and then restarted. The He pressure looks looks low to me 60-70 psi. We may have a leak. HE cryopumps on and pumping the beam lines and tandem. 15L beamline chamber has higher pressure than normal. It is in the mid 10-5 range. There is a mist of oil droplets in the chamber. Could be that the rotary pump burped and oil from the diffusion pump finished up the chamber and the ion gauge is contaminated (best case). Worst case is we have rotary and diff pump oil in the chamber. May need to clean the pump and chamber.
20 Mar 2021 Saturday 20:00 [Ingram] Came in about 7:30 PM. Restarted the LE cryopumps. Sputter source looks like it survived the outage. The HE pumps may have restarted themselves and then shutdown. They need longer to warm up. There is ice on the outside. The LE pumps had no condensation on them. The alpha source vacuum system was off. I tried to restart it but failed. We need to write-up the procedure to restart it. I will come in on Sunday to restart the HE vacuum.
20 Mar 2021 Saturday 18:10 [Gula Hamad] Hello Everyone, There was a power outage today when I entered the lab to refill the HPGe detector and I am not sure how long the power was out. The power came back while I was there and couple of Alarms started . I stopped (reset) the Alarms.
19 Mar 2021 Friday 12:29 [Tom Massey] Pulsing and bunching working! The switch for buncher tubes was in between positions. We now have good signals for pulsing and bunching!!!!
19 Mar 2021 Friday 11:58 [Tom Massey] Started work for pulsing and bunching. We had zero for the buncherf output. Both 24 cm and 8 cm positions were checked. No buncher output was seen in either position. The switch sounded like it was making contact. We stopped the experiment until this is understood.
18 Mar 2021 Thursday 17:00 [carter] BCI counter 10's digit repaired.
18 Mar 2021 Thursday 12:38 [Greg] Fan resecured to process chiller. Shaft key still missing, but set screws will hold it until a key can be found. Everything ready to run.
16 Mar 2021 Tuesday 17:08 [Greg, Don] Impeller to shaft key had sheared off (probably age/corrosion). Mike Myers was able to find a replacement key, nut, and washer, and pump is reassembled using spare pump seal. Pump is running and process chiller has loop temps back to normal. The fan on FC1 on the process chiller is loose on the motor shaft. Will need to repair Thursday.
16 Mar 2021 Tuesday 12:59 [Greg, Don] Investigation of the process water chiller shows that it is not running due to the flow switch not being made. Redundant flow switch also indicating no flow. No pressure change seen at process chiller between motor on and motor off. No current change at motor when closing an isolation valve at the pump. Motor current 4.3A, logbook entry indicates expected motor current is 7.5A. Current working hypothesis is that the pump has failed somehow. Will try to diagnose further after lunch.
15 Mar 2021 Monday 17:50 [Greg] Replacement pump installed for the utility water loop. Everything looks good, need to update the pump model number, and review replacement parts with Doug. Tom noticed the cryo pump on right 45 started making a strange noise. We turned it off, will investigate when time permits. Air/Water alarms started going off this evening. Further investigation shows that the process chiller is calling for all 3 stages to run, but none are running. Will investigate tomorrow.
11 Mar 2021 Thursday 13:21 [Tom Massey] The beam stop in the solid target chamber on the swinger has had a 0.1 mm gold plate added to reduce the gamma-ray background.
03 Mar 2021 Wednesday 15:36 [Tom Massey] Alpha source hasd 750 na DC. Swinger Be target 12 channels 2.4 ns width with 12 nA on target!
03 Mar 2021 Wednesday 15:08 [Tom Massey] Alpha Pulsing and bunching. F/4 had 12 na on target. The gamma-width was 12 channels ~2.4 ns.
02 Mar 2021 Tuesday 16:55 [Greg] Sputter source ion gauge was not working. Replaced gauge, found bad gauge. Replaced gauge again, still not working. Controller has failed. Swapped with Lesker controller that was on 45R, leaving 45R with no ion gauge at the moment. Not sure how to adjust the set points on this controller, or what controller to use on 45R. Cooling water pump is also leaking at the shaft seal. Pump is currently shut off, which allows cooling to function with no load, and reduces the volume of the leak. Can turn the pump back on to run, but needs to be swapped for the replacement, and a new seal ordered so that we can have a backup.
26 Feb 2021 Friday 17:13 [Greg] Doug was able to re-manufacture the utility loop water pump and make it run. Installed it, along with 4 new ball valves, and got the utility cooling loop back up and running. Pressures look good, and I bled some air out of the system. Heat exchanger seems to have brought the temperature down to normal. Replacement pump ordered and arrived today to receiving, should be at the lab on Monday. Will work to get it ready to plug in quickly, should this pump fail or develop a leak.
23 Feb 2021 Tuesday 09:28 [Greg] Found water pump making noise this morning, working to investigate. Facilities electricians and controls guys are here to work on the HVAC systems as well.
17 Feb 2021 Wednesday 09:58 [Justin Warren] Came in this morning (2-17-21 @ 9:30) to check the status of my experiment and both alarm panels were in alarm. On the shutdown module, the Cup In light was on, no lights were on main alarm panel. I reset both systems as per the manual and no further alarms occured.
02 Feb 2021 Tuesday 11:27 [Greg] Backfilled Alphatross with Ar, and found that the orifice was not completely plugged. More investigation found some rubidium compound was built up on the canal holder (part 20) and some obstruction of the canal for the RF source (part 24). The source exit flange (part 21) also appeared to have some buildup of rubidium compounds. I was able to clear out the canal, and poke some of the compounds out of the exit flange, but saw some minor sparking, and decided to stop. Source is reassembled now, and pumping down to high vacuum. Can try to start plasma and produce beam, but the source almost certainly need to be torn down and cleaned.
01 Feb 2021 Monday 14:42 [Greg, Don] Still having problems with alphasource. Probe current limited out at 7.5mA several times, forcing a lower probe voltage. Currently shutting down source (2021-02-01 14:30). Once cooled, we will back-fill with Ar and check for blockage of the orifice between the plasma bottle and the charge exchange chamber. Symptoms seem similar to the last blockage, though not identical.
29 Jan 2021 Friday 14:28 [Greg] Monthly safety inspection found that the emergency backup light in the tunnel had failed. I got the spare from Don and will replace it. I'll put together a procurement request for 1 spare to put on the shelf.
27 Jan 2021 Wednesday 08:56 [Greg] Started Alpha source. Left Probe at 6kV, current was around 3mA. Did not start oven yet, but plasma is running well.
26 Jan 2021 Tuesday 17:02 [carter,Greg] Documenting alphatross heater voltages and currents for proper operation. Heater is 120V 200W. At 200W, 120V would be 1.666A. Resistance = 72ohm. Test values using type K thermocouple to confirm temperature probe... 3.1 mV = 95C, 4.0mV = 120C, 5.2 mV = 146C, 7.4 mV = 175C. 75 volts on the supply voltmeter gives 1.04 Amps on the supply ammeter.
25 Jan 2021 Monday 12:47 [Greg, Don] Power outage recovery this morning. Diffusion pump on 15L beamline was not working. Troubleshooting found failed temp switch; replaced with spare from pump in the LTR. Backing pump for 15L beamline also was not operating. Need to replace circuit breaker. Cryo-pump compressor on 45R beamline was low on UHP He. Warmed, refilled, roughed, and restarted compressor. All vacuum systems now operating properly.
23 Jan 2021 Saturday 16:55 [Ingram] LE and HE extensions pumping down.
23 Jan 2021 Saturday 12:33 [Ingram] Sputter source pumping on its diffusion pump. Alphatross now on turbo, I figured out how to start it. Beamline cryopumps cooling down. 15L beamline diffusion pump cold, target chamber diffusion pump working. Ion source cooling on.
23 Jan 2021 Saturday 10:58 [carter] Checked and restored power to electrical and computer systems after power outage. Restarted accelerator logging and alphatross logging.
23 Jan 2021 Saturday 08:18 [Ingram] Power back about 1:30 AM. Came in about 7:30 AM. Reset all alarms. Started all secondary water pumps. Primary water and HVAC all look good. Turned off beam line cryopumps to regenerate. Will come back later to restart cryopumps. Sputter source backing pump and freon compressor on. Alphatross vacuum power on but I don't know how to turn the turbo on.
21 Jan 2021 Thursday 11:23 [carter, Greg] After alphatross repair last Friday we achieved 550 nA DC alphas from the source yesterday 20 Jan 2021. We noticed reduced RF amplitude on the RF meter so made some adjustments in the alligator clip positions on the tank coil and it increased the RF level back close to the previous value which was very close to the red area on the meter. We moved each alligator clip approximately 0.25" to the right facing the source.
21 Jan 2021 Thursday 11:15 [carter] Added alphatross heater voltage plot to the web based monitoring page.
15 Jan 2021 Friday 16:46 [Greg, Don] TL;DR: Alphasource troubles Thursday and Friday. Some problems found and corrected, will try to run the source on Tuesday. Troubleshooting the Alphasource this morning. After overnight vacuum pumping, vacuum levels in the 10^-8 Torr range were achieved. Don and I isolated the probe voltage terminal, and applied voltage to ground. We read high resistance at the 250V and 500V settings, but on the 1000V setting a plasma was induced in the bottle, glowing purple. Based on the wikipedia entry for gas-discharge lamps, it could be argon, nitrogen, or oxygen. This would seem to match with the symptoms that we experienced yesterday working with Tom and Nisha, in that we created a current limiting discharge somewhere around 1000V. This would also seem to indicate that there is something in the plasma bottle other than vacuum or helium, as vacuum shouldn't arc and the helium supply is off, and the plasma is the wrong color. We also noticed that adjusting the mass flow on the helium did not affect the vacuum reading at the ion gauge. Ran roughing vacuum on plasma bottle, and back-filled source with argon. Removed plasma bottle and found no flow of argon. Used 1.58mm rod that Don obtained to clear out the 2mm diameter opening between the plasma bottle and the cesium vapor chamber, at which point argon was able to flow out. Reinstalled the plasma bottle, disconnected argon, rough pumped and allowed the turbo to pump on the source. Pressure currently 2 x 10^-7 Torr and falling. Megger applied to probe voltage contacts, and previously observed discharge is now gone, indicating improved condition. Plan to allow the vacuum to pump until Monday, when we will try to restart the source again.
08 Jan 2021 Friday 14:54 [Greg] Ran windows update on VESDA system, and paused updates until February. Also managed to fix the audio driver, by rolling back to an older version.
06 Jan 2021 Wednesday 08:50 [Greg] Walkthrough of the lab this morning. Backing pump on the swinger seems louder than I remember. This is the pump that we replaced the bearings and coupler on before break. Did not seem excessively warm, nor did I detect excessive vibration. Will monitor.
23 Dec 2020 Wednesday 10:25 [Ingram] Got 1.2 uA in LE cup of 60 keV 4He- and 1.5 uA in HE cup through the machine at 1MV. Analyszer magnet control not working.
23 Dec 2020 Wednesday 07:48 [Ingram] Alphatross Testing Plasma started at about 5 kV on the probe today, yesterday it started when the rf came on. Oven temp is 165 C and I already have 0.2 uA and increasing. Gas still at .35 sccm. Dropping the flow to 0.2 sccm increased the beam to 0.4 uA, oven now 170 C.
22 Dec 2020 Tuesday 17:24 [Ingram] After source shutdown I went out to check on the 4He MFC. It had readings that make sense now. psia 27 degC 22.6 set pt. 0.000 main display 0.000 SCCM He 0.00 ccm 0.000 sccm I think the 4He MFC had to go through some sort of reset for it to work normally. Go figure. David
22 Dec 2020 Tuesday 16:39 [Ingram] Alphatross testing Gas left running all night, no change in pressure still 3E-5 Torr. Shutdown. Went out to check the MFCs. Only strange thing are some negative numbers for flow on the 4He controller. Noted the 3He MFC had 4 psia. Tried to run 3He MFC. at .35 sccm the pressure went to 2E-5 Torr, looks like the same problem. Tried to run down 3He controller and turn off its flow, but, although it was off, it showed a non zero flow of about 0.15 sccm. Used Run All Down and source off and eventually it went to zero. Tried the 4He MFC again and it was working properly, 0.35 sccm 1.6E-6 Torr. Shutdown and reverted to proper control of the source, CB1 plugged back in etc. Started source and it worked as described in the manual. Ran for several hours at 0.2 uA, after 6 hours refocused and re-tuned magnet and got the beam to 0.65 uA. What worked? Cycling the MFCs and the source controls appeared to fix the problem but why? All this time the turbo has been running at 820 Hz. I kept checking just to make sure that the turbo was not the problem with the pressure.
21 Dec 2020 Monday 10:22 [Ingram] Testing Alphatross. The ion gauge pressure is still reading 3E-5 when the gas starts to flow. The pressure is rising to that level within ten seconds of the source being turned on. The pressure is real and can be seen at the inflection magnet when the gate valve to the source is open - same reading 3E-5. Background pressure at the inflection magnet is 3E-6. Running Alphatross from wall power. The MFC is reading correctly and is set for He. The regulator pressure close to the MFC is reading 10 psi, it was reading 16 psi earlier, probably because the source was not taking any gas and the bottle on the wall is set at about 20 psi. Will leave the gas flowing in this mode for 24 hours, all other supplies are off, CB1 is unplugged. Other - the Alphatross ion gauge on the real time data in the control room is difficult to read owing to some high values last week. Other - facilities just came through to turn the steam/hot water back on in the mechanical room.
18 Dec 2020 Friday 16:26 [Ingram] Shutdown source. The pressure dropped imediately to 2E-7 Torr. It had been falling very slowly but was still 3E-5 Torr before shutdown.
18 Dec 2020 Friday 13:18 [Ingram] Starting Alphatross. IG pressure went to 3.3E-5 Torr when plasma started. Plasma started when RF came on. Will run only RF, magnet and probe until pressure drops to low sixes.
14 Dec 2020 Monday 16:01 [Greg, Devon] TL;DR: Still no vacuum on swinger, turbo might be broken. Swinger work today: replaced ion gauge with the last spare from the shelf. Reinstalled the backing pump. Turbo pump controller is not working, lights just blink, nothing on the LCD, no lights on the turbo pump. Nothing illuminating in the manual, will investigate further as time permits.
11 Dec 2020 Friday 15:20 [carter] Logbook test.
11 Dec 2020 Friday 12:21 [carter] Again, as on 14 Sept 2020, the capacitor in LE tube vacuum gauge was shorted and was replaced. 0.5 uF 4000 VDC. Last spare on the shelf. Used the one salvaged from spare gauge power supply unit that was removed from old duoplasmatron source. Probably only available now from ebay. Checked the supply voltage and it is within ratings, 3600 VDC.
11 Dec 2020 Friday 12:10 [Greg, Devon] Replaced belts on large roughing pump in the vault. Last belt replacement was 2016. Belts were slipping from oil and ragged. Oil level in the sight glass was low, need to think about an oil change soon.
11 Dec 2020 Friday 09:44 [Greg, Devon] Replaced ion gauge on 65R. Roughed out chamber and beam line.
10 Dec 2020 Thursday 12:10 [Devon Jacobs] Came in this morning to find a phase failure alarm. Found many of the usual suspects, ion gauges, gate valves, in an off state. Of particular note was the alphatross vacuum pumping system. We didn't discover the source vacuum was out until much later. The only indication of such a failure is visual inspection of the vacuum control module in the source area, or if we had thought to look at the software which would have been displaying whatever it would with the ion gauge off. Going forward, should consider tapping into vacuum control circuitry to establish an alarm or other kind of indication for a failure. In other news, facility shutdown is progressing. We've also found the process cooling water emergency dump valve to be inop. Solution in work...
04 Dec 2020 Friday 11:34 [carter, leblanc] Compressor blew fuses again. A new set of contacts has been installed. Also a missing bolt was replaced on the solenoid actuator. Hopefully one or the other was the cause of blowing fuses.
02 Dec 2020 Wednesday 16:08 [carter] Test of addition of Greg LeBlanc to logbook "mail to all operators" function.
02 Dec 2020 Wednesday 16:07 [carter] Test of logbook "mail to Greg LeBlanc" addition.
18 Nov 2020 Wednesday 14:37 [carter] The main air compressor went out again today during the run shutting down the machine. Found one blown fuse and one thermal overload tripped again, same as before. This time I have burnished the contacts as this is a possible cause. I will also check on the availability of a new set of contacts for this contactor. Part number is Square D TA81. This set was replaced for the same symptom April 2006.
18 Nov 2020 Wednesday 11:41 [carter] Currently there are software imposed limits for the alphatross controls of the following, Mass Flow for HE3 and HE4 of 0.18 to 0.50 SCCM and also a software limit on the oven voltage of 90 volts max.
18 Nov 2020 Wednesday 11:25 [Tom Massey] Alphatross start up normal. Starting current 1.1-1.2 microamper. This was slighly unstable. Dr. Ingram saw this behavior also but this eventually stablized at the higher value. His test showed that the source mass flow can be reduced to ~0.18 CCM with improved current output. We currently have 200 nA on target with no current on the collimator at the 65R HeBGB target position.
13 Nov 2020 Friday 11:23 [Devon] The beam right object slit is repaired. Took the micrometer from the Y-up slit. Did not have to break vacuum. The failed beam right slit was not at or beyond the zero point, it seemed to be sitting at around the 0.03" open point. But that is only an estimate as to how far open it was. New Beam right zero set when slits were in contact with beam left at zero. The Y-up slit is trapped in place at > 0.4" setting with a hose clamp to prevent the spring from pushing the slit out further.
11 Nov 2020 Wednesday 15:27 [Tom Massey] The neutron sources have been returned to the source room.
08 Nov 2020 Sunday 10:38 [Tom Massey] The beam right object slit no longer moves. Trying to turn by hand is not possible. The shaft does not move at all. We have been switching the object and image slits between 0.030 and 0.200. This is done to determine the energy by the magnet and calibrate the nmr at the smaller setting of 0.030. The slits are then opened to get enough beam to obtain enough counts above background.
07 Nov 2020 Saturday 07:54 [Tom Massey] The alphatross has 1.1 microamps this morning.
06 Nov 2020 Friday 09:14 [Tom Massey] The swinger turbo pump's roughing pump was making loud noises this morning. The turbo pump has been turned off.
04 Nov 2020 Wednesday 16:16 [carter. shafer] At 11:30 PM last evening the main air compressor stopped. Upon inspection found and repaired the following, blown fuse, thermal protector tripped in motor starter relay box, conduit feed termination had worked loose and needed re-connected at motor and a long bolt (1 of 4) that holds the motor bearing housings together had worked loose and needed replacing. Doug had a new piece (1/4 in. all-thread) in the shop to replace the bolt. After the repairs were made the motor started and the compressor is working again.
03 Nov 2020 Tuesday 23:30 [Doug Soltesz] Tonight at 11:20 pm, the "water hot or air low" alarm sounded and the grid voltage quickly increased to around -15 V. The accelerator was immediately shut off and the magnets turned down. We contacted Devon, who said it was probably the air compressor, and he will be in to fix it on Thursday.
03 Nov 2020 Tuesday 23:30 [Doug Soltesz] Tonight at 11:20 pm, the "water hot or air low" alarm sounded and the grid voltage quickly increased to around -15 V. The accelerator was immediately shut off and the magnets turned down. We contacted Devon, who said it was probably the air compressor, and he will be in to fix it on Thursday.
02 Nov 2020 Monday 08:09 [phase failure detected] The phase failure detected alarm on the E2 panel was on this morning. I checked all vac systems and everything seems to be operating normally. Best guess is there was a very short power outage that set off the alarm but did not hurt anything. I will proceed to turning on the source unless I hear otherwise. -KB
30 Oct 2020 Friday 13:31 [carter] Emergency generator oil changed by facilities.
28 Oct 2020 Wednesday 17:40 [carter] The fictitious chain stretch alarm was caused by sparking at the alphatross source. The cause was a diagnostics induced disconnected plasma box fan cable that was not plugged in and hanging from the source platforming sparking to ground. Initial restart of the source produced 450 nA of beam within 15 minutes.
28 Oct 2020 Wednesday 12:48 [Tom Massey] The chain stretch alarm is going off. This happened when the source was being brought up. This came on when the bias voltage was being raised. The alarm was gone when the bias voltage on the alphatross was reduced. When the extractor voltage reached above ~5 kV the alarm would be set off. Sparking???
23 Oct 2020 Friday 15:44 [Tom Massey] The accelerator was conditioned up to 4.5 MV. We are currently running at 3.371 MV.
15 Oct 2020 Thursday 17:15 [Devon] Alphatross update: All 10 g of Rb loaded without issue. Flowed He thru the source over lunch. Then went for operation. Plasma struck right away but vacuum reading was higher than expected as Probe voltage came up. Let it run a couple hours as suggested in the manual, vacuum came down to expected readings. Our instructions have been changed to say the Probe runs at 6 KV instead of 6.5 KV. Not sure how we got to 6.5, but the manual and other labs use 6. Moved on to oven heating and applied voltages. The focus is a key player in tuning now, which we didnít see when the focus element was shorted by Rb. Initially the beam output was less than 100 nA and the gas flow was higher than expected. As time went on a lower gas flow started giving higher beam output, now closer to the expected .2 sccm. I recall seeing this the last time the bottle was changed. The Rb oven behavior is different now. I had used a bit of thermal contact grease on the TC last time. This time I packed the connection heavier with grease. The oven was set to 87.5 V, what had become our standard and gave us a temperature reading in the upper 180s C. We had seen instability appear if the temp reached 190 C around 90 V. After some time of running today it was noticed that the oven temp had already reached 191 C at 87.5 V. The beam had also come up to 250 nA. Lowered the oven volts to get temp back down and then started nursing it back up. We are currently at 185 C at 77.5 V and getting 200 nA. This makes me think that we were not getting good TC readings on the last build and actually over driving the oven. Something had to make the Rb mess. Iím going to leave the source running overnight as it is right now. Perhaps the beam will improve by morning. If or if not, there should be some exploration of new operating parameters on the oven. No idea what the upper limit is at this point. If somebody wishes to explore this tomorrow then great. If not, please shut down the source. Also, the big einzel lens pretty much does nothing. If you raise it high enough it defocuses the beam a little. Another item for further contemplationÖ